The 38003105-1 panel allows mounting of up to four PRL-970-series modules into a 1U rack space. PRL-970-series modules have a nominal height of 1.3" and a nominal width of 2.9". Examples include PRL-414B, PRL-425T, etc.
To mount a module to the panel, remove the 4 screws securing the endplate to the module, then insert the screws through the matching holes in one of the 4 positions in the panel:
On some modules with threaded BNC or Triax connectors, the nut may need to be rotated a few degrees so that the flat edge of the nut is parallel with the opening.
If you need to mount a module such that the DC power input faces the outside of the rack, a PRL-730 Voltage Distribution Module and an 88000102-1 cable may be used to connect power from the inside of the rack, as shown below:
The rear face of the 38003105-1 panel is gold-alodine treated to establish a good ground connection with the enclosures of the mounted modules:
Typical DC resistance between the panel and the module grounds is 0.1 Ohms, although this is not a guaranteed specification.
Cutout detail, 4 places:
Material: 0.12" THK 5052-H32